打開包裝后，檢查設備是否損壞。 根據結構圖，按設備型號安裝設備。 應根據裝箱單檢查配備的部件。 如果加熱方式是電加熱，請根據工作溫度購買相應類型的導熱油（注意：導熱油絕對不允許含有水分）。 添加時，夾套上部的注油口將打開，夾套上部的油將打開。 這位是開放的。 將油注入加油口，等待油位流過。 油位擰緊后，請勿擰緊燃油口以避免壓力。
通過擰緊螺母和螺母來安裝閥門，壓力表和安全閥，以達到密封效果。 兩個接頭的電弧密封表面不得相對于彼此旋轉。 對于所有螺紋聯軸器，應在組裝過程中使用潤滑劑或潤滑油。 石墨避免傷人。 閥門的使用：針閥是密封的，只需要輕輕旋轉閥針，緊蓋可以達到良好的密封性能，防止過大的力，以免損壞密封面。
反應釜壺體和壺蓋的安裝和密封：壺體和壺蓋與墊圈或錐形表面和弧面接觸，主螺母通過按下主螺母擰緊來實現 密封效果好。 擰緊螺母時，須對角對稱。 它應該多次重復擰緊，力應該是均勻的。 不要讓蓋子向一側傾斜以獲得良好的密封效果。 擰緊主螺母時，規定的緊固扭矩不應超過40~120N.M的范圍，以防止密封面被壓壞或過載。 密封表面應特別小心。 每次安裝前，用較軟的紙或布擦拭上下密封表面。 請特別注意不要碰到壺體和蓋子的密封面。 如果合理操作，可以使用10,000次以上。 在密封表面破裂后，需要重新加工以獲得良好的密封性能。 拆卸蓋子時，蓋子應緩慢上下升降，以防止水壺和蓋子之間的密封面相互碰撞。 如果密封件用墊圈（四氟乙烯，鋁墊，銅墊，石棉墊等）密封，則可以通過擰緊主螺母來實現良好的密封效果。
Xiaoyi will introduce to you that the electric heating reactor is installed in the high-pressure operating room which meets the explosion-proof requirements.
After opening the package, check whether the equipment is damaged. According to the structure diagram, install the equipment according to the type of equipment. The fittings should be checked according to the packing list. If the heating mode is electric heating, please purchase the corresponding type of heat conducting oil according to the working temperature (note: heat conducting oil is absolutely not allowed to contain moisture). When added, the oil injection port on the upper part of the jacket will be opened, and the oil on the upper part of the jacket will be opened. This is open. Inject the oil into the filling port and wait for the oil level to flow through. After tightening the oil level, do not tighten the fuel nozzle to avoid pressure.
Valves, pressure gauges and safety valves are installed by tightening nuts and nuts to achieve sealing effect. The arc sealing surfaces of the two joints shall not rotate relative to each other. For all threaded couplings, lubricants or lubricants should be used during assembly. Graphite avoids hurting people. Valve use: Needle valve is sealed, only need to gently rotate the needle, tight cover can achieve good sealing performance, prevent excessive force, so as not to damage the sealing surface.
Installation and sealing of kettle body and lid of reaction kettle: The kettle body and lid contact with gasket or conical surface and arc surface, and the main nut tightens by pressing the main nut to achieve good sealing effect. When tightening nuts, they must be diagonally symmetrical. It should be tightened repeatedly and the force should be uniform. Do not let the lid tilt to one side to achieve good sealing effect. When tightening the main nut, the specified tightening torque should not exceed the range of 40-120 N.M. to prevent the sealing surface from being crushed or overloaded. The sealing surface should be carefully sealed. Before each installation, wipe the upper and lower sealing surfaces with softer paper or cloth. Please take special care not to touch the sealing surface of the pot body and lid. If operated properly, it can be used more than 10,000 times. After the sealing surface is broken, it needs to be reprocessed to obtain good sealing performance. When removing the lid, the lid should be lifted up and down slowly to prevent the sealing surface between the kettle and the lid from colliding with each other. If the seals are sealed with gaskets (tetrafluoroethylene, aluminium gasket, copper gasket, asbestos gasket, etc.), good sealing effect can be achieved by tightening the main nut.